semperguard® nitrile Xpert black
The stylish black colour of the semperguard® nitrile Xpert black, combined with the increased wall thickness, results in a highly resilient glove that is preferred by professional users in various industrial applications. Thanks to the high wall thickness, the glove offers the highest protection of all gloves in the semperguard® disposable glove portfolio and – as the name indicates – is aimed at experts, predominantly in the occupational safety and pharmaceutical sectors. The black colour also makes this glove suitable for additional uses.
Fully textured for optimised grip in damp and wet applications.
Enhanced protection thanks to a significantly higher typical force at break, which has been tested according to EN 455-2.
Single use disposable protective glove, non-sterile
Nitrile butadiene rubber (NBR)
Flat with rolled rim, ambidextrous
Overall length as per EN 455-2
S, M, L, XL, XXL median 240 mm
Measured in double layer in the palm area
min. 0.20 mm
Tightness as per EN 455-1
Force at break
as per EN 455-2
median ≥ 6 Newton
in original packaging if stored as per ISO 2230
Labeling & CONFORMITY TO STANDARDS
EN 420, EN ISO 374-1, EN 374-2, EN 16523-1, EN 374-4, EN ISO 374-5, EN 455-1, EN 455-2, EN 455-3, ISO 2859-1, ASTM D6319, ISO 16604
100 pcs./dispenser S, M, L
90 pcs./dispenser XL, XXL
Failure to observe this information, in particular with regard to chemical resistance, frequency of use and tolerability of the gloves, can result in personal injury and/or material damage. Sempermed is not liable for incorrect use of the gloves nor will Sempermed be liable for improper storage and handling. In case of indications of damage or unusual signs of wear and tear to the gloves, the gloves should not be used. In case of any doubt, expert advice should be sought before use. This information and classification is up to date as of the date of the last update to this page. Technical details are average values from production and may vary in individual cases. This disclaimer is subject to mistakes, printing errors and amendments.
This product contains accelerators of Dithiocarbamate types and Zinc-mercaptobenzothiazol. These products should not be used if hypersensitivity of these substances exists. For further information, a list of substances contained in the glove is available upon request.