sempercare® nitrile sterile
FEEL YOUR STERILE SAFETY: This thick sterile examination glove is made from 100% nitrile. The snythetic, powder-free product is an alternative for people with Type I allergies.
The sempercare® nitrile sterile is the synthetic alternative for people with type I allergies. With regard to sensitivity, fit and grip they are very close to natural latex gloves.
- Good chemical resistance.
- Finger textured for secure grip.
- Suitable for specific cytostatics.
Powder-free sterile examination and disposable protective glove
Nitrile Butadiene Rubber (NBR)
Flat with rolled rim, ambidextrous
Overall length as per EN 455-2
S, M, L, XL median 240 mm
Measured in double layer in the palm area
min. 0.16 mm
typical 0.20 mm
Tightness as per EN 455-1
Force at break
as per EN 455-2
median ≥ 6 Newton
in original packaging if stored as per ISO 2230
Labeling & CONFORMITY TO STANDARDS
EN 420, EN ISO 374-1, EN 374-2, EN 16523-1, EN 374-4, EN ISO 374-5, EN 421, EN 455-1, EN 455-2, EN 455-3, ISO 2859-1, ASTM D6319, ASTM F1671, EN 556-1
Each pair with folded cuff in foil peel pack with lashes for easy handling; tightness statistically checked according to ISO 11607-1
50 pairs / dispenser S, M, L, XL
300 pairs / transport carton S, M, L, XL
*Please note that all Sempermed products must be used in accordance with the Sempermed chemical resistance and the respective declaration of conformity.
Failure to observe this information, in particular with regard to chemical resistance, frequency of use and tolerability of the gloves, can result in personal injury and/or material damage. Semperit is not liable for incorrect use of the gloves nor will Semperit be liable for improper storage and handling. In case of indications of damage or unusual signs of wear and tear to the gloves, the gloves should not be used. In case of any doubt, expert advice should be sought before use. This information and classification is up to date as of the date of the last update to this page. Technical details are average values from production and may vary in individual cases. This disclaimer is subject to mistakes, printing errors and amendments.
This product contains accelerators of Dithiocarbamate types and Zinc-mercaptobenzothiazol. These products should not be used if hypersensitivity of these substances exists. For further information, a list of substances contained in the glove is available upon request.