sempercare® green combines skin protection with environmental protection. The skin-friendly material minimises the risk of contact allergies as no accelerator chemicals are used during production. In addition, the nitrile glove conserves resources through sustainable manufacturing: 12% less water consumption, 17% less CO2 emissions, and no chlorine is used during production. Additionally it is accelerator-free and produced with our innovative green glove technology saving on water and energy consumption. Good for your hands - Good for the planet.
- The skin-friendly material minimises the risk of contact allergies as no accelerator chemicals are used during production.
- More sustainably manufactured: 12% less water consumption, 17% less CO2 emissions, no chlorine used during production.
- Easy donning thanks to synthetic inner coating.
Protects users and patients from the transmission of infections. Glove material is resistant to blood and other body fluids, and protects against bacteria, fungi and viruses including SARS-CoV-2, Hepatitis B & C and HIV.
Powder-free nitrile examination and disposable protective glove, accelerator-free, non-sterile
Nitrile Butadiene Rubber (NBR)
Powder-free polymer coated
Flat with rolled rim, ambidextrous
Overall length as per EN 455-2
XS, S, M, L, XL median 240 mm
Measured in double layer in the palm area
min. 0.12 mm
Tightness as per EN 455-1
Force at break
as per EN 455-2
median ≥ 6 Newton
in original packaging if stored as per ISO 2230
Labeling & CONFORMITY TO STANDARDS
EN 420, EN ISO 374-1, EN 374-2, EN 16523-1, EN 374-4, EN ISO 374-5, EN 421, EN 455-1, EN 455-2, EN 455-3, ISO 2859-1, ASTM D6319, ASTM F1671
200 pcs / dispenser XS, S, M, L
180 pcs / dispenser XL
Failure to observe this information, in particular with regard to chemical resistance, frequency of use and tolerability of the gloves, can result in personal injury and/or material damage. Sempermed is not liable for incorrect use of the gloves nor will Sempermed be liable for improper storage and handling. In case of indications of damage or unusual signs of wear and tear to the gloves, the gloves should not be used. In case of any doubt, expert advice should be sought before use. This information and classification is up to date as of the date of the last update to this page. Technical details are average values from production and may vary in individual cases. This disclaimer is subject to mistakes, printing errors and amendments.