sempercare® cobalt blue
The sempercare® cobalt blue is a lightweight and extremely soft examination and protective glove suitable for various application fields as per intended use. Offering the perfect latex-free alternative for anyone looking for a glove combining exceptional wearing comfort with an indispensable force at break. The soft nitrile formulation wears like a second skin and is especially suitable for people with sensitive skin. The darker striking colour of the glove is not only and eye-catcher, but the use of different glove colours for different applications can help to avoid cross contamination.
- The soft material results in high wearing comfort and a comfortable fit.
- Your daily choice for versatile use.
- Protects the user against chemicals*.
- Protects users and patients from the transmission of infections. The glove material is resistant to blood and other body fluids, and protects against bacteria, fungi and viruses including SARS-CoV-2, Hepatitis B & C and HIV.
Powder-free examination and disposable protective glove, non-sterile
Nitrile Butadiene Rubber (NBR)
Flat with rolled rim, ambidextrous
Overall length as per EN 455-2
XS, S, M, L, XL median 240 mm
Measured in double layer in the palm area
min. 0.10 mm
Tightness as per EN 455-1
Force at break
as per EN 455-2
median ≥ 6 Newton
in original packaging if stored as per ISO 2230
Labeling & CONFORMITY TO STANDARDS
EN 420, EN ISO 374-1, EN 374-2, EN 16523-1, EN 374-4, EN ISO 374-5, EN 421, EN 455-1, EN 455-2, EN 455-3, ISO 2859-1, ASTM D6319, ASTM F1671
100 pcs / dispenser XS, S, M, L
90 pcs / dispenser XL
Failure to observe this information, in particular with regard to chemical resistance, frequency of use and tolerability of the gloves, can result in personal injury and/or material damage. Sempermed is not liable for incorrect use of the gloves nor will Sempermed be liable for improper storage and handling. In case of indications of damage or unusual signs of wear and tear to the gloves, the gloves should not be used. In case of any doubt, expert advice should be sought before use. This information and classification is up to date as of the date of the last update to this page. Technical details are average values from production and may vary in individual cases. This disclaimer is subject to mistakes, printing errors and amendments.
This product contains accelerators of Dithiocarbamate types. These products should not be used if hypersensitivity of these substances exists. For further information, a list of substances contained in the glove is available upon request.