sempercare® silk combines optimal protection from infections with advanced skin care technology. The innovative examination glove boasts a special film enriched with liquid silk on the inside of the glove. This has a positive effect on the skin and optimal protects and nurtures it, also in the case of prolonged wear. The liquid components in the silk lends smoothness to the skin, enabling a pleasant, silky-smooth feeling on the skin. In addition to protection from infections, the comfortable glove offers excellent tactile sense and is also suitable for users and patients with Type-I allergies. Treat yourself to the silk feeling!
- Glove material is resistant to blood and other body fluids, and protects against bacteria, fungi and viruses including SARS-CoV-2, Hepatitis B & C and HIV.
- The liquid silk-enriched inner coating offers a silky-smooth feeling and is suitable for people suffering from Type-I latex allergy.
- The components of the silk inner coating help to protect and nourish the skin, especially when gloves are worn for long periods of time.
Powder-free examination and disposable protective glove, non-sterile
Nitrile Butadiene Rubber (NBR)
Powder-free with silk coating
Flat with rolled rim, ambidextrous
Overall length as per EN 455-2
XS, S, M, L, XL median 240 mm
Measured in double layer in the palm area
min. 0.12 mm
Tightness as per EN 455-1
Force at break
as per EN 455-2
median ≥ 6 Newton
in original packaging if stored as per ISO 2230
Labeling & CONFORMITY TO STANDARDS
EN 420, EN ISO 374-1, EN 374-2, EN 16523-1, EN 374-4, EN ISO 374-5, EN 421, EN 455-1, EN 455-2, EN 455-3, ISO 2859-1, ASTM D6319, ASTM F1671
100 pcs / dispenser XS, S, M, L
90 pcs / dispenser XL
Failure to observe this information, in particular with regard to chemical resistance, frequency of use and tolerability of the gloves, can result in personal injury and/or material damage. Sempermed is not liable for incorrect use of the gloves nor will Sempermed be liable for improper storage and handling. In case of indications of damage or unusual signs of wear and tear to the gloves, the gloves should not be used. In case of any doubt, expert advice should be sought before use. This information and classification is up to date as of the date of the last update to this page. Technical details are average values from production and may vary in individual cases. This disclaimer is subject to mistakes, printing errors and amendments.
This product contains accelerators of Dithiocarbamate types and Zinc-mercaptobenzothiazol. These products should not be used if hypersensitivity of these substances exists. For further information, a list of substances contained in the glove is available upon request.